Electrostatics test explains how you can recognize and detect the effects of electrostatic charges on your own balance. more
Weighing the Right Way
Guidelines for accurate results and better weighing techniques
Factors affecting weighing accuracy can negatively influence the results of the whole analysis, therefore it is important to be aware of the rules that need to be followed when working with laboratory balances.
Balances have developed to the point that, in general, no special weighing room is necessary, except maybe for ultra-micro balances.
Technological advances in electronics have considerably simplified the operation of balances, reducing weighing times. However, this apparent ease of use can lead to insufficient care being taken to avoid the negative influence on analyses results caused be multiple factors affecting weighing accuracy. External influences are defined as physical effects which are measurable for all laboratory balances. Examples of external influences include:
- Electrostatic forces
- Environmental vibrations
- Magnetism effects
- Moisture uptake
The purpose of this guide is to identify most common factors affecting weighing accuracy and to explain the basic rules of weighing the right way with laboratory balances