Moisture determinations need to be carried out reliably and at a speed so that any interventions in the production process can be made quickly. more
To use all functions of this page, please activate cookies in your browser.
Weighing is one of the key activities carried out in every QC laboratory. Usually, it is one of the first parts of a whole analysis chain, such as when a sample or a standard is prepared for HPLC or qNMR analysis. As any weighing errors have the potential to propagate through the whole analysis and deliver an inaccurate final result, the United States Pharmacopeia (USP) has set stringent requirements for balances used for weighing analytes for quantitative measures. A recent update effective August 1, 2019 underscores the need for accuracy and emphasizes the concept of minimum weight as basis for accurate weighing processes.
Fig. 1 Foundation of the United States Pharmacopeial Convention on January 1, 1820 in the Old Senate Chamber of the U.S. Capitol (11 of the 16 delegates – all physicians – present) . The first USP was published on December 21, 1820. Artist: Robert Thom. Painting commissioned by USP, 1957.
The pharmacopeias are central documents for the pharmaceutical industry. These documents are collections of published standards describing requirements for testing chemical and biological drug substances and dosage forms, as well as methods of analysis for medicines. The standards themselves are defined to ensure pharmaceutical products are of the appropriate identity, strength, quality, purity and consistency. While pharmacopeias describe requirements for the testing of drug substances and dosage forms, they do not apply to the manufacturing of the respective products.
One of the oldest and most widely applied pharmacopeias is the U.S. Pharmacopeia (USP) (Fig. 1). The weighing requirements defined in its General Chapter 41 , together with state-of-the-art performance verification activities described in the informational General Chapter 1251 , are quite often regarded as a de facto standard to ensure consistently accurate weighing results in QC across the pharmaceutical industry.
It is important to refer to the USP General Notices and Requirements  to understand the exact scope of General Chapter 41. Section 6.50.20 states (text shortened):
“Solutions for quantitative measures shall be prepared using accurately weighed or accurately measured analytes (see section 8.20).”
Section 8.20 stipulates (text shortened):
“If the measurement is stated to be ‘accurately weighed’, follow the statements in Balances ⟨41⟩.”
With these two requirements, it is evident that whenever a monograph requires material to be “accurately weighed,” the balance on which the weighing process is carried out needs to comply with the requirements described in General Chapter 41.
General Chapter 41 sets three distinct requirements to be applied to materials that must be accurately weighed.
“The weighing shall be performed using a balance that is calibrated over the operating range and meets the requirements defined for repeatability and accuracy.”
Fig. 2 Systematic and random errors of the balance are assessed by specific tests with an acceptance criterion of 0.10 %. The systematic errors have to be tested with a test load between 5 % and 100 % of the balance capacity. The repeatability test load is usually selected at a few percent of the balance capacity.
Two of the three requirements for balances expressed in the above are repeatability and accuracy. By defining specific acceptance criteria for the repeatability and accuracy test, it is ensured that both random and systematic errors of the instrument are minimized. Both acceptance criteria are expressed as relative limit values, 0.10 % (Fig. 2). From a practical perspective, this requirement is quite stringent and ensures that the weighing error is usually small, if not negligible, compared to errors in the subsequent process steps described in the individual monographs when drug substances are tested.
Calibration, the third requirement, is defined in the International Vocabulary of Metrology (VIM) , and it establishes, for weighing instruments, a relationship between the mass value of reference weights and the balance indication including the associated measurement uncertainties.
Of specific practical relevance to balance performance is the repeatability requirement stipulated by USP General Chapter 41 (text shortened):
“Repeatability is assessed by weighing one test weight not less than 10 times. Repeatability is satisfactory if twice the standard deviation of the weighed value, divided by the desired smallest net weight (i.e. the smallest net weight that the users plan to use on that balance), does not exceed 0.10 %.”
A very important consequence of the repeatability requirement is the concept of minimum weight, which has already been described for many years in the informational General Chapter 1251. During the minor revision of USP General Chapter 41 that came into effect August 1, 2019, this description of minimum weight has been moved from General Chapter 1251 to this mandatory chapter. While the repeatability test and assessment have not changed in the revision, the importance of minimum weight for the practical application of the balance in day-to-day use was enforced by this revision.
The new revised General Chapter 41 states:
“The repeatability measurement establishes the smallest net amount of material that may be weighed on the balance in conformance with the 0.10 % limit.”
Based on the repeatability acceptance criterion, (2 x s) / (smallest net weight) ≤ 0.10 %, all masses equal or larger than (2 x s) / (0.10 %) conform with this requirement. The smallest mass that satisfies this criterion is called minimum weight:
mmin = (2 x s) / (0.10 %) = 2,000 x s
Note that the smallest standard deviation (s) that may be used for the assessment against the 0.10 % repeatability requirement and for the calculation of minimum weight is 0.41 d, where d is the digital increment (readability) of the balance. This lower limit is based on the rounding error of a digital indication.
As stated in the repeatability requirement, the smallest net weight is defined as the smallest quantity that the user wants to weigh on the device on a day-to-day basis. This smallest net weight is a user requirement and should not be confused with the minimum weight that is a property of the instrument and which is calculated as described above.
With the two preceding definitions for smallest net weight and minimum weight, the following statement applies: When the smallest net weight (the user wants to weigh) equals or is larger than the minimum weight (as calculated from the repeatability of the balance), then the repeatability criterion of General Chapter 41 is satisfied.
USP General Chapter 1251 provides detailed information regarding qualification and operation of the instrument. While the chapter is specifically written for analytical balances, most of the information presented can also be applied to balances of higher capacity, such as precision balances or bench scales.
Fig. 3 Variability of the minimum weight over time due to changing environmental conditions. To ensure traceability of the weighing results, an as-found calibration is carried out before servicing or adjusting the instrument.
An important part of this chapter is a detailed presentation of state-of-the-art strategies for routine testing of the instrument. The frequency and type of routine testing activities depends on the risk and the required weighing process tolerance. Daily tests are not prescribed, and the usage of built-in adjustment weights in the balance allows the reduction of testing frequency using external weights. However, it is left to the user to specifically define performance verification activities based on the above considerations.
Another important consideration of General Chapter 1251 details the variability of the balance's performance over time triggered by changing environmental conditions and operator influence on weighing accuracy (Fig. 3). To mitigate these influences, it is recommended that the smallest net weight, i.e. the smallest quantity that the operator would like to weigh using the balance, is sufficiently larger than the minimum weight as derived through the repeatability test at a particular moment in time. In other words, applying a safety factor to ensure consistent adherence to the pre-defined weighing tolerance at all times is recommended.
Accurate weighing is key for any quality-relevant weighing application. The USP has defined in its General Chapter 41 clear requirements that should ensure that any weighing application within its scope does not significantly contribute to any overall analytical error. Besides using a calibrated balance, requirements for repeatability and accuracy are established and characterized by an assessment against a specified tolerance (0.10 %). As an important consequence of the repeatability test, the minimum weight can be calculated. This minimum weight describes the smallest amount of net substance that can be weighed while complying with the tolerance requirement.
USP General Chapter 1251 elaborates more fully on routine balance tests by applying a risk-based approach on performance verification activities. Furthermore, it elaborates on the variability of the balance's performance over time, which leads to the concept of the safety factor.
Applying the requirements of USP General Chapter 41 and best practices from General Chapter 1251 ensures that the user can rely on the accuracy of the instrument over time.
Category: Pharmaceutical Quality Control | Compliance
 USP General Chapter 41 “Balances”, United States Pharmacopeial Convention, Rockville, USA, USP42-NF37 Supplement 1, August 2019
 USP General Chapter 1251 “Weighing on an Analytical Balance”, United States Pharmacopeial Convention, Rockville, USA, USP42-NF37 Supplement 1, August 2019
 Tirumalai, G., Long, A. “United States Pharmacopeial Convention: Respecting the past, moving confidently into the future”, United States Pharmacopeial Convention, Rockville, USA, retrieved from https://www.usp.org/sites/default/files/usp/document/about/usp-history.pdf, accessed on 2019 Aug 9
 USP General Notices and Requirements, United States Pharmacopeial Convention, Rockville, USA, USP42-NF37 Supplement 1, August 2019
 JCGM 200 (VIM), “International Vocabulary of Metrology – Basic and General Concepts and Associated Terms”, 3rd ed., JCGM, 2010, https://www.bipm.org/utils/common/documents/jcgm/JCGM_200_2012.pdf, accessed on 2019 Aug 9
Moisture determinations need to be carried out reliably and at a speed so that any interventions in the production process can be made quickly. more
Routine testing of laboratory balances by the user is essential to uncover malfunctions early and eliminate inaccuracies. more
Shop floor board meetings are key elements of successful lean management principles. A shop floor approach is relevant not only for production environments. more
These extremely durable balances are designed to withstand accidental overload, harsh chemicals, dust and dirt more
The combination of reliable weighing performance and ergonomic operation makes the XSR analytical balances for the strong routinier, even under harsh conditions under the fume hood more
With the model XPR6UD5, Mettler-Toledo presents the first ultra-micro balance on the market, delivers unique 0.5μl resolution more
Balance accessories for your laboratory balance can help to simplify your processes and ensure you get the accurate results you want. more
METTLER TOLEDO has launched its new Dubai Free Zone Competence Center. This new facility will offer advanced analytical support for all industries operating in the Middle East and North Africa. It will also provide hands-on equipment training for METTLER TOLEDO’s range of products, which ar ... more
Mettler Toledo presents a special offer for 2016 - a limited-edition, full-color lab calendar. More than a way to track days and dates, this 12-month calendar will provide a series of enlightening topics devoted to ensuring weighing accuracy in all critical lab processes. Using calendar pag ... more
The English version of the q&more online portal is now available online. The q&more portal was developed in a unique cooperation between succidia AG and CHEMIE.DE Information Service GmbH, and is supported by METTLER TOLEDO. It is aimed specifically at decision-makers, and uses attractive c ... more
For about 200 years, any mass given in kilograms, grams and micrograms referred to a defined artifact accessible in only a single location on earth. A small metal cylinder had represented one kilogram since the metric system was introduced in France. Its successor became world famous as the ... more
Easy access and navigation of historical measurement data stored in central laboratory software systems is important to support the documentation and, in particular, the traceability of measurement results. For some instruments it is also useful to grant users access to past measurement data. more
The process industry is seeing manufacturing structures change, production chains becoming global and higher requirements placed on manufacturing processes. Businesses need to adapt to declining sales volumes and the demand for more specific product characteristics. more
Christian Müller-Schöll, born in 1966, studied mechanical engineering at the University of Stuttgart, Germany, and graduated as Diploma-Engineer. At the ZHAW Zurich University of Applied Sciences he obtained a certificate for Integrated Risk Management. He headed the primary mass calibratio ... more
Igor Knapp, born in 1968, is a qualified computer science engineer and Team Leader for Interaction Design at Mettler-Toledo GmbH. In 1993, he successfully completed a design education at the Basel School of Design (Switzerland) and subsequently worked for more than seven years in corporate ... more
Wolfgang Boos, born in 1965, initially trained as an Information Technician before studying Technical Computer Science at Flensburg University of Applied Sciences (Germany), graduating with a diploma degree. He subsequently worked as a software developer at the Forschungszentrum Jülich, Ins ... more
The 1 kg redefinition is part of the biggest overhaul of the International System of Units (SI) since its inception in 1960. During the last several years, Mettler-Toledo has been privileged to participate in several strategic partnerships that have contributed to the newly established and ... more
XPR Analytical Balances are the right choice for challenging weighing applications, such as weighing the smallest sample sizes, with exceptional accuracy and a readability of 5 micrograms. Save time, money and materials, and invest in a balance that consistently delivers results you can tru ... more
In today's high pressure research and manufacturing environments, having the right equipment is key to achieving quality and efficiency. With customers demanding short lead times and reliable delivery dates, you need a balance you can trust to deliver accurate results - and which is conveni ... more
USP Chapter 41 weighing requirements are mandatory in a Pharmaceutical Quality Control laboratory, where weighing is a fundamental step. (…)
The q&more concept is to increase the visibility of recent research and innovative solutions, and support the exchange of knowledge. In the broad spectrum of subjects covered, the focus is on achieving maximum quality in highly innovative sectors. As a modern knowledge platform, q&more offers market participants one-of-a-kind networking opportunities. Cutting-edge research is presented by authors of international repute. Attractively presented in a high-quality context, and published in German and English, the original articles introduce new concepts and highlight unconventional solution strategies.
> more about q&more
q&more is supported by: