Mandatory requirement for quality control of medicinal products in the European Member States who exported to the European market more
To use all functions of this page, please activate cookies in your browser.
Weighing is one of the key activities carried out in every quality control (QC) laboratory. Usually it is one of the first steps in a whole analysis chain, e.g. when a sample or a standard is prepared for subsequent dilution and HPLC or qNMR analysis.
For many decades the United States Pharmacopeia (USP) has set stringent requirements for balances used for weighing analytes for quantitative measures in its General Chapter 41 “Balances” [1, 2]. These requirements ensure that weighing errors are small or even negligible for the result of the analysis. Other Pharmacopoeias have not previously set similar requirements. This situation will change from July 2021, when a General Chapter for balances is published in the European Pharmacopoeia (Ph. Eur.) . The European Pharmacopoeia is the single reference compendium for the quality control of medicines in 39 European countries . It is legally binding for pharmaceutical companies that intend to bring pharmaceutical products onto the market in one of the 39 countries, and is therefore not only relevant for European pharmaceutical companies but for pharmaceutical companies in other regions of the world which intend to export to Europe. The new General Chapter 2.1.7 “Balances for Analytical Purposes” stipulates requirements that are very closely aligned with the respective requirements of the USP. The new chapter is mandatory for any weighing procedure described in a Ph. Eur. monograph from January 1, 2022 and legally enforceable from that date onwards.
Within its Equipment Performance section, General Chapter 2.1.7 sets three distinct requirements for balances used to weigh materials according to Ph. Eur. monographs:
i) Balances must be periodically calibrated.
Between calibrations, performance checks must be carried out periodically to assess the
ii) random error (precision) and
iii) systematic error (accuracy) of the balances.
Before analyzing in detail the three requirements on calibration, precision and accuracy, it is important to understand how accuracy and precision are defined in pharmaceutical regulations. Accuracy is defined in ICH Q2(R1)  and describes the systematic error of measurements. This concept is called “trueness” in all other fields of metrology, and defined as such in the International Vocabulary of Metrology (VIM) . Precision is defined similarly in ICH Q2(R1) and the VIM, and describes the random error of measurements.
By defining specific acceptance criteria for both performance checks, it is ensured that the random and systematic errors of an instrument are minimized. Both acceptance criteria are expressed as relative limit values of 0.10 %.
It is worthwhile noting that the performance requirements set in Ph. Eur. General Chapter 2.1.7 are commensurate with those in USP General Chapter 41.
According to the General Chapter 2.1.7, balances need to be calibrated periodically. Calibration covers a set of activities carried out on a measuring instrument to understand its behavior. This is done by establishing a relationship between known values (measurement standards) and the associated measured values (indications) of the instrument being calibrated. The chapter stresses the importance of including a statement of measurement uncertainty that allows metrological traceability of the measurement results.
Besides being calibrated, an instrument can also be adjusted. When adjusting an instrument, its indications are modified in a way as to correspond – as far as possible – to the values of the measurement standards applied. General Chapter 2.1.7 stresses the importance of performing calibration before (“as found”) and after (“as left”) an adjustment or any significant operation on the balance, for example upon repair or transfer to another location. This concept ensures the traceability and validity of measurement results performed on the balance during routine use.
After discussing the requirement of calibration, the following paragraphs will focus on the requirements of the two performance checks concerning precision and accuracy that must be carried out periodically between calibrations.
A. Repeatability requirement – minimum weight
Ph. Eur. General Chapter 2.1.7 requires a repeatability test to be carried out with no less than 10 replicate measurements. The repeatability is satisfactory if ( 2 × s / msnw ) × 100 ≤ 0,10, with s being the standard deviation of the indicated values, and msnw the smallest net weight defined by the user as the smallest net amount of substance that will be weighed on the balance. If s < 0.41 × d , in which d is the readability (scale interval) of the balance, s is replaced by 0.41 × d for the assessment.
A very important consequence of the repeatability requirement is the concept of minimum weight. The minimum weight is the smallest net sample mass that can be weighed on the balance, whilst continuing to comply with the repeatability test criterion. All masses equal or larger than (2 × s × 100) / 0.10 conform to this requirement, and the smallest mass that satisfies this criterion is called minimum weight:
mmin = (2 × s × 100) / 0,10 = 2,000 × s
When calculating the minimum weight, s is replaced by 0.41 × d if s < 0.41 × d.
While the smallest net weight is a user requirement and defined as the smallest quantity that the user wants to weigh on the device on a day-to-day basis, it should not be confused with the minimum weight that is a property of the instrument and is calculated as described above. With these two definitions, the following statement applies: when the smallest net weight (that the user wants to weigh) equals or is larger than the minimum weight (as calculated from the repeatability of the balance), then the repeatability criterion of General Chapter 2.1.7 is satisfied.
Fig. 1 Test weight selection for repeatability testing. The test weight does not need to be at the working point of the balance. Ph. Eur. Chapter 2.1.7 recommends choosing a test weight that is 5 % of the balance capacity or slightly below. This is consistent with the recommendations of USP General Chapter 1251 .
For the repeatability test, General Chapter 2.1.7 suggests taking a test load of no more than 5 percent of the nominal capacity of the balance, so a 10 g load for a balance with a nominal capacity of 200 g would be appropriate (see Fig. 1). A test load that is sufficiently larger than the expected minimum weight is easier to handle and avoids human errors being introduced during the repeatability test that could negatively influence the assessment of the instrument's performance.
B. Accuracy Requirement
Further to repeatability, Ph. Eur. General Chapter 2.1.7 stipulates how to assess the accuracy of the balance. Three major weighing parameters may affect the accuracy of the balance: sensitivity, linearity and eccentricity. The weighing parameter that most significantly affects accuracy is sensitivity. Sensitivity characterizes the change of indication of the balance divided by the change in load, so a balance without sensitivity deviation would indicate exactly the weighing value that corresponds to an applied test load at nominal capacity . As sensitivity is the weighing parameter that most significantly affects the accuracy of the balance, it is considered sufficient to only assess sensitivity when carrying out performance checks. However, the acceptance criterion for sensitivity is set to 0.05 %, i.e. half of the overall accuracy acceptance criterion of 0.10 %. This takes into consideration that the other weighing parameters also influencing the accuracy of the balance might also contribute to the systematic error. All added up must not jeopardize the overall accuracy requirement of 0.10 %.
Fig. 2 Test weight selection for accuracy testing. The test weight must be between 5 % and 100 % of the balance’s capacity.
It needs to be noted that at the lower end of an analytical balance's measurement range, repeatability is the most dominant contributing factor to balance errors. Therefore, it is not meaningful to assess accuracy with a small test weight, as a potential systematic error would be masked entirely by the limited precision of the instrument, expressed by its repeatability. To avoid using small loads for testing systematic deviations, Ph. Eur. General Chapter 2.1.7 stipulates that the mass of the test load must be at least 5 % of the balance's capacity (see Fig. 2).
C. Usage of built-in adjustment weights
In addition to testing weighing instruments with external weights, it is accepted practice to adjust the instruments by means of built-in reference weights. Such practice allows reducing the frequency of sensitivity tests with external reference weights [7, 9, 10], and is also described in Ph. Eur. General Chapter 2.1.7.
At this point, it is important to point out a very frequent misconception that has prevailed in the industry for decades. Almost everybody in the pharmaceutical industry who works in quality control speaks of a “daily balance test” which assesses the accuracy of a balance. For electronic balances with a built-in weight, a daily test of sensitivity with an external reference weight is an outdated practice that does not need to be performed. Ph. Eur. General Chapter 2.1.7 focuses on a testing approach where it is the quality management system of the user that defines the frequencies of calibration and the individual performance checks.
Accurate weighing is key for any quality relevant weighing application. The European Pharmacopoeia has defined in its General Chapter 2.1.7 clear requirements to ensure that any weighing application within a Ph. Eur. monograph does not significantly contribute to the overall error of the analysis. Besides using a calibrated balance, requirements to ensure repeatability and accuracy are established and characterized by an assessment against a specified tolerance of 0.10%. As an important consequence of the repeatability test, the minimum weight is calculated. It establishes the smallest amount of net substance that must be weighed in order to comply with the permitted tolerance. The weighing requirements of Ph. Eur. General Chapter 2.1.7 are commensurate with the requirements set in USP General Chapter 41. Therefore, the European Pharmacopoeia has made an important step towards the international harmonization of weighing requirements between different pharmaceutical compendia.
Category: Pharmaceutical Quality Control | Compliance
 USP General Chapter 41 “Balances”, United States Pharmacopeial Convention, Rockville, USA, USP-NF 2021, 2021
 Fritsch, K. (2019) Weighing in the US Pharmacopeia – Minimum Weight and Routine Testing According General Chapters 41 and 1251, 2019 Aug 26, https://q-more.chemeurope.com/q-more-articles/296/weiging-in-the-us-pharmacopeia.html (Ger.: Die US-Pharmakopöe auf der Waagschale – Mindesteinwaage und Routineprüfung gemäß den USP-Kapiteln 41 und 1251, 2019 Aug 26, https://q-more.chemie.de/q-more-artikel/296/die-us-pharmakopoee-auf-der-waagschale.html)
 Ph. Eur. 10.6, General Chapter 2.1.7 “Balances for Analytical Purposes”, EDQM Council of Europe, Strasbourg, France, European Pharmacopoeia (Ph. Eur.) 10th Edition, Supplement 10.6., July 2021
 European Directorate for the Quality of Medicines & HealthCare (EDQM), https://www.edqm.eu/en/news/european-pharmacopoeia-supplement-106-now-available, accessed on 2021 May 11
 ICH Q2 (R1), “Validation of Analytical Procedures: Text and Methodology”, International Conference on Harmonization of Technical Requirements for Registration of Pharmaceuticals for Human Use, 2005
 JCGM 200 (VIM), “International Vocabulary of Metrology – Basic and General Concepts and Associated Terms”, 3rd ed., JCGM, 2010, https://www.bipm.org/utils/common/documents/jcgm/JCGM_200_2012.pdf, accessed on 2021 May 11
 USP General Chapter 1251 “Weighing on an Analytical Balance”, United States Pharmacopeial Convention, Rockville, USA, USP-NF 2021, 2021
 R. Nater, A. Reichmuth, R. Schwartz, M. Borys, P. Zervos, “Dictionary of Weighing Terms - A Guide to the Terminology of Weighing”, Springer Berlin Heidelberg, 2009, ISBN 978-3-64202013-1
 Questions and Answers to Current Good Manufacturing Practices, Good Guidance Practices, Level 2 Guidance – Equipment, U.S. Food and Drug Administration, Rockville, USA, retrieved from http://www.fda.gov/drugs/guidancecomplianceregulatoryinformation/guidances/ucm124777.htm
 LAB 14, “Guidance on the calibration of weighing machines used in testing and calibration laboratories”, 6th ed., United Kingdom Accreditation Service UKAS, Feltham, UK, 2019
Mandatory requirement for quality control of medicinal products in the European Member States who exported to the European market more
This application note explains how the XPR Automatic Balance can significantly reduce exposure risk during automated powder and liquid dispensing. more
The method presented enables using less substance and solvent, ultimately significantly saving time and money more
50 % fewer OOS – less rework ✓ Economical use of rare or expensive samples with 30 % lower minimum weight ✓ Maximum user safety by automated dispensing ✓ more
Proper verification and documentation assists you to fulfill the demands of industry regulations and company audits more
METTLER TOLEDO has launched its new Dubai Free Zone Competence Center. This new facility will offer advanced analytical support for all industries operating in the Middle East and North Africa. It will also provide hands-on equipment training for METTLER TOLEDO’s range of products, which ar ... more
Mettler Toledo presents a special offer for 2016 - a limited-edition, full-color lab calendar. More than a way to track days and dates, this 12-month calendar will provide a series of enlightening topics devoted to ensuring weighing accuracy in all critical lab processes. Using calendar pag ... more
The English version of the q&more online portal is now available online. The q&more portal was developed in a unique cooperation between succidia AG and CHEMIE.DE Information Service GmbH, and is supported by METTLER TOLEDO. It is aimed specifically at decision-makers, and uses attractive c ... more
Metrological traceability is necessary for consistent, comparable and valid measurement values. As a fundamental concept of modern-day metrology, it allows measurement results to be internationally recognized and is required by norms and accreditation standards all over the world. The trace ... more
Lean management has expanded in scope to establish itself in corporate functions beyond production. In present-day laboratory operations, it forms a basis for optimized processes and efficient quality management, providing valuable tools for the digital transition to Laboratory 4.0. more
Weighing is one of the key activities carried out in every QC laboratory. Usually, it is one of the first parts of a whole analysis chain. As any weighing errors have the potential to propagate through the whole analysis and deliver an inaccurate final result, the United States Pharmacopeia ... more
Igor Knapp, born in 1968, is a qualified computer science engineer and Team Leader for Interaction Design at Mettler-Toledo GmbH. In 1993, he successfully completed a design education at the Basel School of Design (Switzerland) and subsequently worked for more than seven years in corporate ... more
Christoph Jansen, born in 1962, studied chemistry at the University of Cologne and received his doctorate there from D.O. Hummel in instrumental analytics. He was a founding member of the association “Angewandte Spektrometrie Köln e.V.” (Applied Spectrometry, Cologne) and worked as an appli ... more
Daniel Fuchs, born in 1969, studied Services Marketing and Management (MAS SMM) at the University of Lucerne and holds degrees in Product and Service Marketing of the Swiss Federal government. At the Zurich University of Applied Sciences he gained the certification of Lean Manager Service f ... more
Mandatory requirement for quality control of medicinal products in the European Member States who exported to the European market (…)
The q&more concept is to increase the visibility of recent research and innovative solutions, and support the exchange of knowledge. In the broad spectrum of subjects covered, the focus is on achieving maximum quality in highly innovative sectors. As a modern knowledge platform, q&more offers market participants one-of-a-kind networking opportunities. Cutting-edge research is presented by authors of international repute. Attractively presented in a high-quality context, and published in German and English, the original articles introduce new concepts and highlight unconventional solution strategies.
> more about q&more
q&more is supported by: